Pipeteador Drummond Pipet-Aid XP
Gama formada por 4 modelos :
- Pipet-Aid XP2: diseño ergonómico; permite una manipulación cómoda; gracias al nuevo cargador, se carga durante la manipulación; bomba ultrasilenciosa; se suministra con 4 filtros adicionales, soporte y cargador ;
- Pipet-Aid: es un instrumento fundamental, ligero, autónomo e inalámbrico ;
- Pipet-Aid XP: más rápido, más compacto y con más funciones, suministrado con cargador, Suministrado con soporte;
- Pipet-Aid XP Gravity: igual que Pipet-Aid XP, pero con una función de gravedad (distribución por gravedad, sin empuje). Sistema gravity, útil en el cultivo celular para evitar separar las capas de células.
Pipeteador Drummond AID XP
Specifically designed to minimize operator fatigue, the Portable Pipet Aid XP is smaller and lightweight. It accepts most pipettes from 1 to 100 mL.
Each pipette filler and pipet dispenser button has three speeds, (F) Fast, (M) Medium, (S) Slow. The medium speed is suitable for most routine techniques. Fast speed can deliver up to 50 mL in 5 seconds and can be used for rapid filling of large pipets or breaking up cell clumps. Slow speed is ideal when using small pipettes or when more precise delivery is required. A gravity drain model is available which substitutes a gravity drain (G) function for the fast (F) mode on the dispense button.
The Portable Pipet Aid XP is powered by a 9 volt (Ni-MH) rechargeable battery.
A Tissue Culture Nosepiece, which features a unique filter element, will cease to allow air to pass in either direction if it becomes wet, alerting the user to possible contamination of the rubber insert.
Speed Controls: The XP has three settings on each button to set the desired speed range (S=slow, M=medium, F=fast). The fill and empty buttons are independent of each other and can be set at different speeds. The speeds are changed by simply rotating the button until a detent is felt and the letter is upright when held in the right hand. The buttons are also proportional in nature. As with all other Pipet-Aids, the further you depress the button, faster filling or dispensing will result. With both buttons set at “M”, the XP will operate at the same speed as the Portable Pipet-Aid. Rotating the button to “S” will reduce the speed to approximately half of the medium speed while rotating to “F” will double the medium speed. DO NOT CHANGE SPEED RANGES WITH LIQUID IN THE PIPETTE.
Changing the Filter & Nosepiece Removal (View a video on the “Maintenance Video” tab): Unlike other Pipet-Aids with the T.C. nosepiece, the XP has a one piece nosepiece that contains the filter and rubber insert. This nosepiece locks into place and will not loosen when removing the pipette. To remove nosepiece, slide the nosepiece release button towards the handle and turn the nosepiece one quarter turn counter-clockwise.
Battery Replacement (View a video on the “Maintenance Video” tab): The battery is contained in the upper portion of the handle. Removal of one phillips screw (B), will enable you to remove the battery compartment door which will give you access to the battery. Use only Drummond replacement batteries (4-000-035). Under normal conditions, these batteries should last approximately 2 years.
Charging the Battery: The Drummond replacement battery is a Ni-MH battery which should be charged overnight following use. No “memory” will be developed and overcharging is not a problem.
Use of the Removable Stand: This lightweight stand slides on to the top of the XP enabling the user to set the Pipet-Aid XP down without removing or emptying the pipette. The stand will prevent liquid from running back into the nosepiece and prevent the pipette tip from contacting anything thus maintaining sterility. When not needed, the stand is easily removed.